A multimeter, also known as an electrical tester, circuit meter, or circuit multimeter, is a single-unit measuring instrument that can measure multiple electrical values such as voltage, resistance, and current during experiments and evaluations. They also aid in the maintenance and inspection of electrical equipment.
Although there are some differences depending on the industry or type of business, those with an analog meter display are usually called "testers" or "analog multimeters," while those with a digital meter display are called "multimeters".
Additionally, large-sized meters that are used in a stationary position, such as on a desk, are sometimes distinguished as digital multimeters (DMMs).
Although they are referred to in several ways, their basic functions remain the same.
- (1) Test probe(probe pen/test lead)
This is a physical input device to connect with the target circuit or device under test to take the measurements.
- (2) Measurement terminal
This receives signal input from the test probe. The position of the terminal where the test probe is inserted differs depending on the values to be measured.
- (3) Function switch (range switching switch)
This switches the measurement modes, such as AC voltage, DC voltage, current, and the measurement range.
- (4) Function button (select button)
This is used to activate ancillary functions. In a digital multimeter, the function switch and function button are combined to switch the measurement mode. For, example, press the function button to switch on the LCD display.
- (5) Zero ohms adjuster knob
This knob is used to correct the display variation caused by internal resistance, etc.
- (6) Meter Scale
- (7) Meter pointer
- (8) Display board
Let us review the various measurement methods.
First, in an analog multimeter, adjust the zero position of the multimeter as a preliminary preparation before taking measurements, and also check the continuity. Set the range switch to the resistance measurement mode, and place the test lead pins against each other; if the value is close to 0Ω, it is OK.
Measurements with a multimeter are essentially as follows:
(1) Insert the red plug of the test lead into the + input terminal and the black plug into the - input terminal.
(2) Adjust the measurement range.
(3) Make the test pin contact the measurement area.
(4) Check the scale that matches the measuring range of the switching knob from the meter scale, and read the numerical value shown by the ammeter's pointer (in the case of a digital multimeter, read the numerical value from the display panel).
Thus, the measurement is performed in four steps. Excessive current or voltage may cause a malfunction. Therefore, if the measurement value is unpredictable, start the measurement from the maximum range in any mode and then lower the range while observing the situation.
How to measure AC/DC voltage
After inserting the test lead into the terminals, switch the range knob to AC/DC voltage mode to select the expected voltage range. Then, place the black test probe on the "-" side of the measurement target and the red test probe on the "+" side.
How to measure resistance
After inserting the test lead into the terminals and switching the range switch knob to resistance mode, apply the test sticks to each other and turn the zero-ohm switch knob to set the pointer to the 0 Ω position.
After alignment, place the red and black test pin on both ends of the measurement target. Be sure to turn off the power to the measurement object before taking the measurement.
How to measure DC current
After inserting the test lead into the terminals and switching the range switch to DC mode, place the red test stick on the + side of the measurement target and the black test stick on the - side. Note that the analog multimeter cannot measure AC.
There are two types of multimeters: analog and digital., Although digital multimeters have been more prevalent as the world is becoming digitalized, both have their own merits and demerits.
Advantages of digital multimeters
- Because it displays numerical values, there is no variation among multimeters.
- The multimeter itself is highly accurate and can take accurate measurements.
- Various functions can be accessed by switching modes, such as temperature measurement mode using thermocouples and current-checking mode.
- Easier to operate than analog multimeters.
- Wider range of products. digital multimeters range from inexpensive hand-held types to costly high-performance ones, which are safer, more durable, and more accurate than cheaper multimeters. Other features include simultaneous measurements of multiple channels, a wide range of currents and voltages, and USB connectivity.
Disadvantages of digital multimeters
- There are numerous functions; it takes time to familiarize.
- Reading fluctuating values is difficult.
- Operation requires batteries
Advantages of analog multimeters
- Easier to understand by sense.
- Easier to recognize fluctuating values such as voltage and current because the needle can be shaken.
- • Can measure voltage and current without batteries.
Disadvantages of analog multimeters
- Variations in values due to internal resistance resulting in measurement errors.
- Requirement to read the scale.
Related Technical Articles
Matsusada Precision manufactures and sells a wide range of power supply equipment, including DC power supplies, high-voltage power supplies, bipolar power supplies, electronic loads, and AC power sources used for electric circuit experiments.
Palm-sized DC Power Supply
Ultra compact and high-performance DC power supply with a sufficient output power of 36W in "palm size".
High Resolution DC Power Supply
Low noise DC power supply with digital control and series regulator system
Compact DC Electronic Load
Compact tabletop electronic loads, ideal for circuit evaluation