Connection of load
- Please use a short lead wire that is sufficiently thick for the connection.
- Please use PVC electric cable (105°C) that can fully tolerate the voltage used. It is necessary to consider current capacity, length limit of output wire by sensing (0.5 V/lead) and so on for wiring with load. Please refer to the following diagram to determine the thickness of cable.
|AWG||mm2||Max current (A)|
Use several cables or copper bar for model over 350 A.
Parallel connection of load
A power supply has no direct branching, but the load is branched using cables.
The following definitions shall basically take precedence. In case there is any specific definition described in other parts of the brochure, the definition has higher priority than the following ones. The specification in this brochure indicates the value at rated power output (full scale) shown after a two-hour warm up.
Scope of Specifications
The following specifications are applied within a range of 10 to 100% of the rated output.
- Temperature coefficient
- Linearity of output
- Linearity of monitor
- Linearity of indications
Displays rms including high-frequency noise.
The preset value does not accurately show the actual status of output. When accurate setting is required, actually output with no load and set the voltage. Also, set the current setting to the desired value by outputting with the output terminal shorted and gradually increasing current.